Title & Document Type: A 50 MHz Programmable Pulse Generator This manual may contain references to HP or Hewlett-Packard. Hp A Pdf User Manuals. View online or download Hp A Operating, Programming And Servicing Manual. View and Download HP A operating, programming and servicing manual online. 50 MHz programmable. A Pulse Generator pdf manual download.
|Published (Last):||9 September 2017|
|PDF File Size:||13.44 Mb|
|ePub File Size:||17.96 Mb|
|Price:||Free* [*Free Regsitration Required]|
Typical signals in External Burst mode The item you’ve selected was not added to your cart.
HP 8112A Manuals
Width Control Verification Test The standard control board is divided into the following areas: In all cases, the bit is set to 1 to indicate that the condition described is true. No additional manuql charges at delivery! This effect is particularly usefull for fast ranging between decades. The offset signal is applied to the non-inverting input at the junction gp CR, CR to ensure a constant reference point.
Marvin Test Solutions, a vertically-integrated aerospace test and measurement company, has created and delivered innovative, reliable test systems for factory, depot, intermediate, and flightline use since Refer to Chapter 4 Operating for information on the trigger controls. Set up the HP 8 A as follows: Refer to Chapter 4 Operating for more information on modulating the output signal. Normally its output is negative because its inverting input is at a higher potential 5.
The U pulse key gives an additional pulse in Gate and External Burst modes. If re-adjustment is necessary, repeat Fixed, Linear and Gauss adjustments as required.
One non accessible location for currently active mode and parameter settings.
Typical Pulse Display Typical output in Normal mode I’m voting to close this question as off-topic because no effort whatsoever is shown. The right hand vernier key is used to alter the value of X, to determine which file the parameter set will be stored as.
Trending Price New. Period Performance Test 1.
Variable Transitions To make operation flexible, variable transitions rise and fall times can be programmed from 5 ns to ms. Minimum delay is 75 ns and a maximum delay of 1 second is obtainable.
8112A 50 MHz Pulse Generator
UlO, Ull schematic B B Use the vernier keys to change the address. It is recommended to separate the commands with a comma, because this makes it easier to read such a programming message.
The examples list the basic operating steps in the order in which they would normally occur after switching on. These components are summarised in Table Width finally reaches the value of 2 ms. It is held tightly by the RFI seals at the front of manua frame.
Connect the equipment as shown in the setup figure 2. Microprocessor board architecture Offset-vernier Control-voltages HP manusl The time needed, varies with the type and number of the commands sent.
This item will ship to United Statesbut the seller up not specified shipping options. Same as triggered mode except pulses are output for the duration of the gated signal.
Microprocessor Board A3 – Schematic 3 Set the HP A a. Repeat last two steps and re-adjust if necessary.
Our service centers may be able to perform calibration if no repair parts are needed, but no other support from Agilent is available. Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – mqnual in a new window or tab. And if you can’t do something, it’s a good indication to check the capability of the device.
HP A Pulse Generator Operating & SERVICE Manual | eBay
Error recognition and prompt helps the operator to recover from incorrect front panel or programming operation. Multiple Commands You can send multiple programming commands on the same line. Verify that the output signal is inverted by pressing the [coMPLl key.
Each of the bits 0 thru 5 summarizes a speciflc type of error. Set amnual the HP A as follows: Remove the case by sliding it backwards.